Visit us on the web at for more information: • Price Quotations • Drivers· Technical Specifications. Manuals and Documentation . Model System DMM Instruction Manual Rev. D. Manual Type: Primary User; Part Number: D. Release Date: Sep 17 View and Download Keithley DMM instruction manual online. 5 function autoranging digital multimeter. DMM Measuring Instruments pdf manual.
|Published (Last):||27 September 2013|
|PDF File Size:||5.6 Mb|
|ePub File Size:||1.11 Mb|
|Price:||Free* [*Free Regsitration Required]|
Keep in mind that each function can have its own unique filter value. The U3 value will not be cleared when read; thus, the U3 value keithly always current. To store one reading in the buffer after each press of the ENTER button, send T7 one-shot on external trig- ger over the bus.
The current LO limit will be displayed. This feature eliminates errors due to a low level external voltage 916 configured in series with the unknown resistor. Dangerous arcing can also result when the meter is set to a voltage manal if the minimum voltage spacing is reduced.
For 4-terminai measurements connect the resistance to the instrument as shown in Figure The controller sets ATN false. Cursor location is indicated by the bright, flashing display digit.
keithly Bit 5 Error — Set when one of the following errors have occurred: This command will clear all bits in the SRQ mask. The entire command string is ignored when a conflict error occurs.
In the continuous trigger mode, the instrument takes a continuous series of readings. LOCAL Note that the instrument goes into the local mode, and that operation of the front panel mwnual has now been restored. If the value of Y is larger than can be handled by the particular range, the overrange message will be displayed, indicating the instrument must be swit- ched to a higher range.
196 System DMM Instruction Manual | Keithley
If it is desired to modify the value of M, do so using the data keys. Transmission of U4, U5 and U6 will not occur until the buf- 169 is full.
Model instruction manual with schematics, Rev. Connect the test leads to the input of the Model and short them together. These commands are covered in detail in the following paragraphs. In order to send a device-dependent command, the con- troller must perform the following steps: This program is useful when slope calculations are required for a series of measurements. Mxnual the data store is full.
Has external cal enable switch on rear. Calibration standard for next units: Each press of one of these buttons toggles the display between the two available conditions. The instrument will return to the previous operating state with the zero modifier enabled. Configure kfithley storage interval and buffer size of the data store by sending the appropriate Qn and I commands over the bus see paragraph 3.
Thus the dynamic measurement range of the instrument is To speed the response to large step changes, the Model digital filter employs a “window” around the displayed average. Upon power up, the instrument will do a number of tests on itself.
When the filter is enabled, readings will be filtered before msnual displayed. Noise and Source Resistance— The limit of sensitivity in measuring voltages with the Model is determined by the noise present. The format of UO status is shown in Figure The last two statements acquire data from the instrument and maual the reading string on the CRT. Mail will not be published required. These can be large compared to the signal which the Model can measure.
Maanual zero and leave it disabled for the remainder of the DCV verification procedure.
System DMM Instruction Manual | Keithley – TechRecovery, LLC
These user programmed conditions will then replace the previously saved default conditions on power up. C Controller — The Model does not have controller capabilities. The values of M and B can be changed by utilizing Pro- gram Repeat the procedure for each of the ranges with manuql voltages.