View and Download Hioki BT instruction manual online. BATTERY TESTER . BT Test Equipment pdf manual download. Also for: Bt View and Download Hioki instruction manual online. BATTERY HiTESTER. Test Equipment pdf manual download. comparator tables and send them back to the Pencil The enhanced measurement current in the . HIOKI (Shanghai) Sales & Trading Co., Ltd.: .
|Published (Last):||12 June 2009|
|PDF File Size:||19.68 Mb|
|ePub File Size:||8.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
Data stored to the selected memory no.
operations manual hioki 3554
Example 2 When the measured resistance value exceeds the resistance upper limit no. Press key three times. This will turn off the disconnect-detection feature.
In this example, unit C has Press keys to select been selected The actual time needed may vary depending on the processing speed of the personal computer used. Disconnect the test lead from the battery subject to measurement.
BATTERY TESTER BT – Hioki
Be sure to connect the test lead to the terminals at a fixed location. Page 17 Measurement Precautions Observe the following to avoid electric shock and short circuits. Use the measurement values to judge hiokki wear. When using this averaging feature, icon is displayed on the screen.
Hioki BT3554 Instruction Manual
Bluetooth label only for BT p. Operating Precautions Operating Precautions Operating Precautions Follow these precautions to ensure safe operation and to obtain the full benefits of the 355 functions. Probing locations when the subject of measurement has width and thickness values.
Frequently Asked Questions 9. Press key to select resistance range. Primarily for use with lead acid batteries, large angle of probe application, 2. When the distance between the terminals on the battery subject to measurement is greater than the distance between the holes on the zero-adjust board, use the holes in the corners for the zero-adjust feature. If the measured object exhibits other electrical characteristics in addition to resistance, as shown in the below image, we can hioi the synchronous detection system to obtain the hiokki resistance of the object.
When the following message has been displayed, check the USB cable connection or the power supply and then click [OK]. DC resistance will not equal effective resistance the real part of impedance; displayed on the instrument.
operations manual hioki |
When the following message has been displayed, check the USB cable connection or the power supply and then click [OK]. Up to comparator conditions can be set.
If you have installed Application Software for in another directory, enter the path to that directory instead of the one shown in the example above. Disconnect the test lead manaul the battery subject to measurement. Press the 0ADJ key for at least 2 seconds.
However, note that the following yioki will not be cleared: But for high levels, the induced voltage distorts the signals, causing incorrect synchronous detection. Starting The Software 7. Refer to “Section 1.
Press key to turn the power off. The instrument will detect the mxnual sensor and display the measured temperature automatically. Using the zero-adjust feature is recommended prior to measurement. A number from 1 to can be selected. After replacing the fuse, replace the fuse cover and screws before using the instrument.
Turning On The Comparator 4.
Measurement ceases when in PC mode. Used for deleting stored measurement values. Displayed when the auto-memory feature is enable. Indicates that descriptive information is provided manial.